The A function generator and all associated options will be The Keysight Technologies A func- . Operating manual, service manual, quick. Service Guide. Publication Number (order as manual set) The Agilent Technologies A is a 20 MHz synthesized function. Buy Keysight Technologies A Function Generator 20MHz GPIB, LAN, USB A. Operating manual, service manual, quick reference guide, IntuiLink.
|Published (Last):||14 September 2018|
|PDF File Size:||11.55 Mb|
|ePub File Size:||13.41 Mb|
|Price:||Free* [*Free Regsitration Required]|
Agilent Technologies 33220A User Manual
You may want to disable the front-panel beeper for certain applications. Used in the internal triggered burst mode only. Also, a remote front panel interface is provided, which allows you to control the instrument over the network.
Burst Chapter 7 Tutorial Burst connector can be configured as an output to enable the A to trigger other instruments at the same time as its manial trigger occurs.
Each character must be a letter “a” Press the Shape softkey to select the shape of the modulating waveform. The default is INT. Nonvolatile arb waveform memory corruption detected The non-volatile memory used to store arbitrary waveforms has detected a checksum error.
AGILENT TECHNOLOGIES A USER MANUAL Pdf Download.
UNIT command on page for more information. Chapter 5 Error Messages Arbitrary Waveform Errors Not able to delete a built-in arb waveform You cannot delete any of the five built-in waveforms: List the names of all waveforms currently available for selection.
However, waveforms of greater than 16, 16K points cannot be edited from the Agilent A front panel. Symmetry represents the amount of time per cycle that the ramp wave is rising assuming that the waveform polarity is not inverted.
The factory setting is enabled ON. The rear-panel 10 MHz In and 10 MHz Out connectors present only if Option is installed allow synchronization between multiple Agilent A function generators see connection diagram below or to an external 10 MHz clock signal. The default is 10 Hz.
Chapter 7 Tutorial Creating Arbitrary Waveforms For example, consider an arbitrary waveform consisting of 10 cycles of a sine waveform. The commands used for FSK are listed on the next page. 332220a 3 Features and Functions Pulse Width Manuaal PWM Width Deviation The width deviation represents the variation in width in seconds in the modulated waveform from the width of the original pulse waveform.
Select the source from which the function generator will accept a trigger. Page 97 Chapter 3 Features and Functions Pulse Width Modulation PWM Width Deviation The width deviation represents the variation in width in seconds in the modulated waveform from the width of the original pulse waveform.
All characters are converted to upper case. Chapter 3 Features and Functions Output Mannual Output Frequency As shown below, the output frequency range depends on the function currently selected.
33220A Function / Arbitrary Waveform Generator, 20 MHz
Page Chapter 3 Features and Functions System-Related Operations Display Control For security reasons, or to speed up the rate at which the function generator can execute commands from the remote interface, you may want to turn off the front-panel display. To print the manual completely, please, download it. You can create an arbitrary waveform from the front panel as described in the following section, or you can use the Agilent IntuiLink software provided on the CD-ROM included with the Agilent A.
This chapter does not give a detailed description of every front-panel key or menu operation.
Sweep Overview The following is an overview of the steps required to generate a sweep. Output Configuration Chapter 3 Features and Functions Output Configuration Output Configuration This section contains information to help you configure the function generator for outputting waveforms. Option is the External Timebase Reference, which provides rear panel connectors 10 MHz In and 10 MHz Out and circuitry to allow synchronization between multiple Agilent A function generators or to an external 10 MHz clock signal.
Duty cycle represents the amount of time per cycle that the square wave is at a high level assuming that the waveform polarity is not inverted. The specified waveform data is divided into samples such that one waveform cycle exactly fills waveform memory see the illustration below for a sine wave.
Check the list of parameters to verify that you have used a valid parameter type. Page Chapter 5 Error Messages Execution Errors Data out of range; large period limits minimum pulse width For a pulse waveform, the minimum pulse width is 20 ns for periods up to 10 seconds. If a command requires more than one parameter, you must separate adjacent parameters using a comma as shown below: This generic message indicates that the offset voltage has been limited to an upper or lower boundary. System-related Operations Chapter 3 Features and Functions System-Related Operations System-Related Operations This section gives information on topics such as instrument state storage, power-down recall, error conditions, self test, and front-panel display control.